What should electromagnetic flowmeter pay attention to when measuring medium with bubble noise
Firstly, meet the length requirements of the upstream straight pipe section of the electromagnetic flowmeter, standardize the installation of the instrument, select the installation site away from the heat source, reasonably use the pipeline flow rate, and select PFA fluoroplastic lining with high finish and high-purity alumina industrial ceramic conduit. These measures will help to prevent or reduce the occurrence of vortex and gas separation. In other words, it is possible to avoid problems by improving the sensor manufacturing process, improving the environmental conditions and installation conditions of the instrument, and installing an exhaust valve upstream of the instrument.
Secondly, setting the damping time and function of electromagnetic flowmeter reasonably can also solve the false alarm of bubble noise measurement. The damping time is selected according to the pulse width of bubble noise in the flow signal. Generally, the damping time should be 3 ~ 5 times of the pulse width of bubble noise. If the pulse width of bubble noise is 10s, the damping time shall be 30 ~ 50s. The specific selection shall be based on the required control accuracy. The control error of 3 times of pulse width is 5%, and the control accuracy of 5 times of pulse width is higher than 1%.
Increasing the damping time of the instrument can effectively solve the impact of this pulse bubble noise, but also bring the disadvantage of slow response, that is, when the real flow fluctuates, the response of the instrument is very slow. This is undoubtedly a difficult problem for the cooling water system requiring sensitive control. In order to solve this problem, the intelligent electromagnetic flowmeter can use software logic judgment, that is, the method of gross error processing. In case of such failure, it is determined whether the flow changes or the bubbles rub the electrode by adjusting the insensitive time of the flow and the limit of the change range. If it is not the noise of bubbles rubbing the electrode, the CPU performs normal sampling, operation and digital filtering; If it is determined that bubble noise is generated, cut off the measured value and maintain the previous flow measurement value. In this way, the damping time during normal flow measurement is still 3 ~ 6S. Only when there is bubble noise, the insensitive time and the system control time will be prolonged according to the length of pulse width setting.
When we reasonably select the change rate limit value and insensitive time value of the electromagnetic flow converter with coarse error suppression function, the converter can not only suppress the false alarm caused by bubble noise, but also maintain the set damping time value when the instrument works normally.
No.5, Shenzhen Avenue, Huanglong Industrial Park, Kaifeng, Henan, China
Copyright© Kaifeng Meihe Automation Instrument Co., Ltd.
Copyright© Kaifeng Meihe Automation Instrument Co., Ltd. All Rights Reserved
Powered by : www.300.cn